Only for production companies: VNA autocal-box: RapACal NA-RC1
We offer the hole production datas with one working solution of RapACal for NVA-companies.
Our product RapACal allows you to perform an easy and rapid calibration with your VNA. It supports:
- One to n-port network analyzer,
- Accurate calibrations from DC to 18 GHz,
- Supports N- or 3.5mm-connectors,
- Simple purpose through a graphical user interface,
- Support VNA error correction.
Rapid Automatic multi-port Calibration box (RapACal) allows an easy and rapid one- or multi-port (n-port, n = 1, 2, 3, …) calibration of Vector Network Analyzers (VNA) up to 18 GHz. The calibration is quite simple. The users just need to make „n+1“ connections to calibrate n-ports. The number of ports is unlimited. RapACal will be controlled over VNA-internal PC interface (LAN and USB) with a graphical user interface from a windows operating system (NT, XP, Vista).
The necessary architecture of RapACal to calibrate a multi-port network analyzer is quite simple.
Figure 2 shows the RapACal-hardware.
The graphical user interface (running in the VNA) is shown in Fig. 3.
For older VNAs, the software can be installed on a external PC. This solution NA-RC1 do not support multi-port analyzers with a switching network for port extension. On request HHF support also these multi-port VNAs.
The performance of RapACal is demonstrated as following. The ripple test for error-estimation is presented in Fig. 4. The DUT is a Rosenberger high-end 50 ohm air line, terminated with a short. The ripple delivers a source match of the system better than 40 dB.
We offer a system composed of
- Hardware box with N (prefered) or 3.5mm,
- Software and operation manual.
- Complete specifications and detailed applications: RapACal NA-RC1
We offer the hole production datas, the software and hardware information with one working solution of RapACal for NVA-companies.
More information are available from Info@HHFT.de.